Schedule Maintenance Of The Glove Box
Etelux reminds you that in addition to the normal inspection of the glove box, the day-to-day servicing work of the glovebox is also essential
one. In advance of and following every single do the job, first clean up the ecosystem to make the ground dry and clear. Then clean the devices, which includes the interior and outer surfaces of the Acrylic Glove Box box, the observation process, the operate surface, etcetera., to be no cost of debris and clear.
2. Thoroughly document the h2o, oxygen information, and gas usage of the products in the glove box every single day to stay away from insufficient fuel force and lead to the machines to quit jogging.
3. Pay back awareness to the quantity of fuel in the working fuel cylinder at any time and replace it in time.
4. In case of electrical power failure, press the "vacuum pump" button on the panel to make the vacuum pump run right after power on
five. Observe: Following electric power on, if there is a difficulty that it are unable to begin, the probability is that the vacuum pump energy cable is loosely related. The second rationale is that the vacuum pump’s have change is not turned on
6. When you want to set the sample into the glove box, use a tiny changeover chamber as considerably as attainable. Right before opening the changeover door, fork out awareness to no matter whether the force gauge pointer on the valve cavity is at the zero placement. If it is, you can open up it directly if not, you will need to convert the swap underneath the valve cavity to the "filling" place to enable the gas enter the glove box changeover compartment, and then switch the switch back to the intermediate posture, then you can open up the changeover compartment Set the sample in the doorway, and then near the doorway. When inserting samples, be thorough not to touch the interior wall of the transition chamber with your hands, put on gloves to pick and position the sample
Executing these schedule routine maintenance tasks can lessen the reduction charge of the glove box, lengthen the provider daily life of the glove box, and also lessen the occurrence of mishaps